Köhler Illumination: Interactive Java Tutorials
Transmitted Light Microscopy
This tutorial explores the optical pathways in a transmitted light microscope. Instructions and a discussion about how to operate this tutorial appear immediately below the window.
To operate the tutorial, use the slider bars beneath the microscope to control the aperture of the field diaphragm (left slider), the aperture (condenser) diaphragm (middle slider), and the illumination intensity (right slider). The button controls whether the internal ("inside") or external ("outside") view of the microscope is selected.
Mortimer Abramowitz - Olympus America, Inc., Two Corporate Center Drive., Melville, New York, 11747.
Matthew J. Parry-Hill, John C. Long, and Michael W. Davidson - National High Magnetic Field Laboratory, 1800 East Paul Dirac Dr., The Florida State University, Tallahassee, Florida, 32310.
BACK TO ANATOMY OF THE MICROSCOPE
Questions or comments? Send us an email.
© 1998-2013 by
Michael W. Davidson and The Florida State University.
All Rights Reserved. No images, graphics, scripts, or applets may be reproduced or used in any manner without permission from the copyright holders. Use of this website means you agree to all of the Legal Terms and Conditions set forth by the owners.
Last modification: Wednesday, Mar 26, 2014 at 02:23 PM
Access Count Since July 30, 1998: 83689
For more information on microscope manufacturers,
use the buttons below to navigate to their websites: