Köhler Illumination: Interactive Java Tutorials
At Low Magnifications
Microscopy at magnifications below 5x is done either with the swing-out condenser lens in the "out" position or with a specialized condenser capable of being used only with low magnification objectives.
This tutorial explores utilization of the field diaphragm to control contrast in low magnification microscopy, as opposed to the condenser aperture diaphragm, which is used for contrast adjustment with high magnification objectives.
Mortimer Abramowitz - Olympus America, Inc., Two Corporate Center Drive., Melville, New York, 11747.
Matthew J. Parry-Hill and Michael W. Davidson - National High Magnetic Field Laboratory, 1800 East Paul Dirac Dr., The Florida State University, Tallahassee, Florida, 32310.
BACK TO ANATOMY OF THE MICROSCOPE
Questions or comments? Send us an email.
© 1998-2015 by
Michael W. Davidson and The Florida State University.
All Rights Reserved. No images, graphics, scripts, or applets may be reproduced or used in any manner without permission from the copyright holders. Use of this website means you agree to all of the Legal Terms and Conditions set forth by the owners.
Last modification: Friday, Nov 13, 2015 at 01:19 PM
Access Count Since April 30, 1999: 25884
For more information on microscope manufacturers,
use the buttons below to navigate to their websites: